SILVER SPRING, Md., Nov. 14, 2018 /PRNewswire/ -- The U.S. Food and Drug Administration today alerted health care providers and patients about the serious complications that can occur when using medications not approved for use with implanted pumps that deliver medication into the spinal fluid to treat or manage pain. Complications may include dosing errors, pump failure, opioid withdrawal, infection and other complications like pain, fever, vomiting, muscle spasms, cognitive changes, weakness and cardiac or respiratory distress.
"The treatment of pain has become increasingly complex. While medical devices, such as implanted pumps that deliver medication directly into the spinal fluid, have the potential to play an important role in treating pain, their use must be judicious and their instructions for use must be carefully followed. This is especially true when it comes to implantable pumps that deliver analgesic medicine directly into the nervous system," said FDA Commissioner Scott Gottlieb, M.D. "We urge health care providers, patients and caregivers to be aware of the information the FDA is providing today so they can make informed treatment decisions. The FDA will continue to monitor the safety of these and other medical devices involved in the treatment of pain, and take action where needed to protect patients."
Implanted pumps are surgically inserted under the skin and connected to an implanted catheter to deliver medicines. They are used to treat diseases and conditions, including pain, by delivering medicine into the body, including directly into the spinal fluid, or intrathecal space. Implanted pumps require a health care provider to periodically refill the pump with medication. Intrathecal delivery of medication with an implanted pump for pain management is generally reserved for patients who are not adequately managed by other pain treatments.
Currently, the FDA-approved implanted pump labeling (Instructions for Use) identifies pain medicines that have been evaluated by the FDA for compatibility with the pump, including Infumorph and Prialt. However, not all pumps are currently approved for use with Prialt. Drugs approved for intrathecal administration must meet additional safety standards over those administered in other ways because the spinal cord and brain tissue are highly sensitive to preservatives or infectious organisms such as bacteria or viruses, which can lead to serious complications.
The FDA has found that patients are sometimes being treated with medications that are not approved (including compounded medicines as well as hydromorphone, bupivacaine, fentanyl and clonidine) for use with an intrathecal implanted pump. The reasons sometimes cited for the use of drugs not approved for use with implanted pumps have included that approved medications listed in the pump labeling may have undesirable side effects or may fail to deliver adequate pain relief for some patients. The FDA is alerting providers that using drugs not approved for use with these pumps may lead to serious risks to patients due to pump failure or dosage errors.
The FDA has determined that implantable intrathecal pump failure is more frequent with the use of medicines not approved for use with the pump. For example, some medicines or fluids may contain preservatives or other characteristics that can damage the pump tubing or lead to corrosion of the pumping mechanism. This may cause the implanted pump to perform in unexpected ways, including motor stalls, which ultimately stop the medication delivery, leading to potential opioid withdrawal. When a pump fails, patients must undergo surgery to remove or replace the pump.
Dosage errors may also occur when medicines that are not approved for intrathecal administration are nonetheless used in these pumps. Programmable implanted pumps have dose calculation software that provides options for users to select pre-programmed medicines and concentrations identified in the approved pump labeling to help prevent unintended dosing errors. The accuracy of the software calculations depends on using the approved medicine, medicine concentration and medicine characteristics. For example, if there is more than one medicine in the pump reservoir, the pump software can only calculate the dose based on the infusion rate of a single medicine.
The FDA recommends that health care providers review the implanted pump labeling to identify the medicines and medicine concentrations approved for use with the specific pump.
The FDA issued this safety communication after reviewing medical device reports, premarket device applications, mandated FDA post approval studies, publicly available scientific literature, current device labeling, and information from health care providers and device manufacturers.
Patients and providers, who experience an adverse event due to an implanted pump or suspect an infusion pump is having problems, are encouraged to file a voluntary report through MedWatch, the FDA Safety Information and Adverse Event Reporting program.
For more information:
- Use Caution with Implanted Pumps for Intrathecal Administration of Medicines for Pain Management: FDA Safety Communication
The FDA, an agency within the U.S. Department of Health and Human Services, protects the public health by assuring the safety, effectiveness, and security of human and veterinary drugs, vaccines and other biological products for human use, and medical devices. The agency also is responsible for the safety and security of our nation's food supply, cosmetics, dietary supplements, products that give off electronic radiation, and for regulating tobacco products.
Media Inquiries: Alison Hunt, 240-402-0764, firstname.lastname@example.org
Consumer Inquiries: 888-INFO-FDA
SOURCE U.S. Food and Drug Administration